Calibrations in the Metering Research Facility (MRF) typically consist of multiple flow rates (6 to 10) at a single pressure and temperature. At each flow rate, six consecutive 90-second runs are performed to verify meter repeatability. Because critical flow nozzles are used for flow control and measurement, there are discrete flow rates available in either loop based on the nozzle combination.
Flow points specific to your test conditions can be estimated by using the online flow calculator. A standard calibration generally takes two days—the first day for installing the meter, related piping, and secondary instrumentation (pressure and temperature transmitters supplied by the MRF) and the second day for the flow test.
If you have specific testing/calibration needs, please contact the MRF.
Standard flow meter calibrations are typically performed in one of the two legs of the HPL test section or in one of the parallel branches of the LPL. Special piping arrangements to replicate field installations can also be accommodated in the facilities. The test ranges available are shown in the following table.
|Maximum Flow Rate, MSCFH||7,100*||663|
|Maximum Flow Rate, ACFH||82,800*||45,000|
|Pressure, psig||150 to 1,100||50 to 200|
|Temperature, Gas °F||40 to 150||40 to 150|
|Pipe Size, inch||2 to 20||1 to 8|
|Specific Gravity**||0.55 to 0.97||0.55 to 0.97|
|Test Fluid||Natural Gas/N2||NG/N2|
|Primary Flow Standard||Weigh Tank||Weigh Tank|
*Maximum flow rate with sonic nozzle flow control. Higher flow rates are available, but without sonic nozzle flow control.
**The typical specific gravity of natural gas at the MRF is approximately 0.57
|Flow Rate||0.1 to 0.25% (R)|
|Pressure||0.015% (F) of value|
|Temperature||0.05 °F (F)|
|Specific Gravity||0.05 (R)|
|Reynolds No||1.0% (R)|
*R = Reading
F = Full Scale